TensCare have many years experience in providing a modern , safe drug-free solution for labour pain and Perfect Mama Tens is our latest TENS maternity unit in our maternity range.
Our TENS maternity unit uses the latest technology and is, above all, portable and easy to use. Perfect mamaTENS will help you eliminate the anxiety that comes with labour as it is fully safe for both, the mother and the baby.
There is a lot of scientific and medical evidence behind effectivity of TENS in alleviating childbirth pain. Keeping that in mind, we have designed our TENS maternity unit Perfect mamaTENS to not only tick all the boxes but go an extra mile. It is effective not just during the birth but also helps with back pains during pregnancy and can help with a quicker postpartum recovery.
Our drug-free childbirth pain relief TENS unit is approved by the NHS, as well as its use being proven scientifically. Additionally, as a safety feature, it has an open circuit system integrated, which means if the electrode does not have full contact, the power will be cut off automatically.
Please note: rental options are not available for this product.
Do NOT use if you have a pacemaker (or if you have a heart rhythm problem) or with any electronic medical devices. Using this unit with electronic medical devices may cause erroneous operation of the device. Stimulation in the direct vicinity of an implanted device may affect some models.
Do NOT use during the first three months of pregnancy. It is not known whether TENS may affect foetal development.
Do NOT use on the abdomen in the later stages of pregnancy. Stop using immediately if you experience unexpected contractions.
Do NOT use when driving, operating machinery, or similar actions needing muscular control. Loose electrode pads, damaged leads, or sudden changes in contact may cause brief involuntary muscle movements.
Do NOT use to mask or relieve undiagnosed pain. This may delay diagnosis of a progressive condition.
Do NOT use if you have, in the area being treated: active or suspected cancer or undiagnosed pain with a history of cancer. Stimulation directly through a confirmed or suspected malignancy should be avoided as it may stimulate growth and promote spread of cancer cells.
Do NOT use electrodes on the front of the neck. Stimulation on the front of the neck can affect your heart rate or cause contraction of the throat.
Do NOT use electrodes across the chest. Very strong stimulation across the chest may cause an extra heartbeat and/or rhythm disturbances to your heart, which could be lethal.
Do NOT use Perfect mamaTENS while simultaneously connected to high frequency surgical equipment as it may result in burns at the site of stimulator electrodes and possible damage to the stimulator.
Do NOT use Perfect mamaTENS in close proximity (e.g. 1 m) to a shortwave or microwave as this may produce instability in the stimulator output.
Do NOT use electrodes near the thorax as this may increase the risk of cardiac fibrillation.
Caution should be used if you have a bleeding disorder as stimulation may increase blood flow to the stimulated region.
Caution should be used if you have suspected or diagnosed epilepsy as electrical stimulation may affect seizure threshold.
Caution should be observed when using the device at the same time as being connected to monitoring equipment with body worn electrode pads.
TENS machines may affect ECG based foetal heart rate monitors, causing them to give a very high reading. Please advise hospital staff that you are using a TENS device.
Caution should be used following recent surgical procedures. Stimulation may disrupt the healing process.
Caution: Not intended for use in an oxygen rich environment.
Caution: Not intended for use in conjunction with flammable anaesthetics or flammable agents.
Caution: The patient is an intended operator.
Caution: Do not service and maintain the device while in use.
Caution: Maintenance and all repairs should only be carried out by an authorized agency. The manufacturer will not be held responsible for the results of maintenance or repairs by unauthorized persons.
If necessary, we will provide circuit diagrams, component part lists or other information that will assist authorized service personnel to repair the device.
Caution: The operator should not touch the patient at the same time when touching the battery output.
Caution: Keep away from pets and pests
Caution: Do not permit use by persons unable to understand the instructions such as those with cognitive disabilities, i.e.; Alzheimer’s disease or dementia.
Caution: Keep away from children under 5 years of age. Long cord - risk of strangulation in infants.
Caution: should be observed when using the Perfect mamaTENS at high strength settings. Prolonged use at high settings may cause muscle injury or tissue inflammation. The Perfect mamaTENS is capable of delivering outputs in excess of 10 mA (RMS) or 10 V averaged over any period of 1 sec.
DO NOT PLACE ELECTRODE PADS:
On skin, which does not have normal sensation. If the skin is numb too great a strength may be used, which could result in skin inflammation.
On broken skin. The electrode pads could encourage infection.
On the front of the neck. This could cause the airway to close, giving
breathing problems. May cause sudden drop in blood pressure (vasovagal response).
Over the eyes. May affect eyesight or cause headaches.
Across the front of the head. Effect on patients who have had strokes or seizures is not known. May affect your sense of balance. The effects of stimulation on the brain are unknown.
ELECTRODE PADS CAUTION:
Caution: Do not ignore any allergic reaction to the electrode pads: If a skin irritation develops, stop using TENS, as this type of electrodes may not be suitable for you. Alternative electrode pads specially made for sensitive skin are available.
Caution: Do not use this device with leads or electrode pads other than those recommended by the manufacturer. Performance may vary from specification. Electrodes with smaller surface area may cause tissue irritation.
Caution: Do not use high intensity settings if electrodes are smaller than 50x50mm.
Note: Electrode pads supplied have no measurable latex content.
TO KEEP YOUR DEVICE IN GOOD WORKING ORDER, OBSERVE THE FOLLOWING ADDITIONAL CAUTIONS:
Caution: Do not immerse your device in water or place it close to excessive heat such as a fireplace or radiant heater or sources of high humidity such as a nebulizer or kettle as this may cause it to cease to operate correctly.
Caution: Keep the device away from sunlight, as long-term exposure to sunlight may affect the rubber causing it to become less elastic and crack.
Caution: Keep the device away from lint and dust, as long-term exposure to lint or dust may affect the sockets or cause the battery connector to develop a bad contact.
Caution: Temperature & Relative Humidity of operating: 5°C to +40°C, 15% to 93% R.H. Temperature & Relative Humidity of storage: -25°C to +70°C, up to 93% R.H.
Caution: There are no serviceable user parts. Do not attempt to open or modify the TENS unit. This may affect the safe operation of the unit and will invalidate the warranty.